1]S.Times (Switch Times): It is calculated by number of times you have been switched on the machine
2]O.P.Mpa (Operating pressure): It shows the operating pressure in the system
3]O.Time (Operating Time): It is calculated by minutes, refer to present running time
4]Acc.Time (Accumulated Running Time): It will display Total number of hours that oxygen machine was operated
1]After starting the medical oxygen concentrator, the LCD screen is lighted, and it will return to screen saver mode in 15 minutes. But if you press the right key during working, the screen will be lit again.
a. P.O.: Power switch (green lamp): If an oxygen machine is connected with electricity, the green light will glow. *Note: This is the only light that will glow for this model while operating the machine, all other lights that mentioned below, will glow when there is any issue with the machine.
b. P.F.: Power failure(red lamp): It means power failure alarm, when no electricity, or any loose contact with power cord or power switch or fuse is damaged, the red light will turn on an audible alarm meanwhile. Make sure to check all connections are good.
c. L.P.: Low Pressure(Yellow lamp): Low Pressure alarm means when the inside system pressure is below 0.12, the light will be red and you may hear a continuous alarm, meaning there is some air leakage in the machine.
d. It is mandatory to use 1 KV voltage stabiliser to maintain life of the medical Oxygen Concentrator. We do not recommend the inverter, as the machine performance might get affected, so please make sure to use 1 KV stabiliser. And always have a backup cylinder handy depending on your needs.
The two buttons are used for timing adjustment, and each press of the left button(△) will increase timing by 10 min, the max timing is 40 hours. And each press of the right button(▽) will decrease timing by 10min. When the right button(▽) is pressed to reduce timing till “0”, the medical oxygen concentrator will turn off automatically when you set some desired amount of time by using these Timing buttons.
Humidifier which is used for humidifying oxygen and preventing throat (preventing Dryness of throat) and nasal mucosa stimulated by dry oxygen and dry hard sputum difficult to spit out. Always make sure to maintain water level up to Half of the bottle not more than half. If water is more than half level, water might get inside of the cannula, which is not advisable.
1]Model: GVS 5L Oxy-pure
2]Flow Rate: 0.5-5 L/ min
3]Oxygen Concentration: 93% (±3%)
5]Sound Level: ≤ 43 DBA
6]Power: AC 230/50
7]Power Consumption: 300 W
8]Rated Power(VA): 540
9]Weight: 16 kgs
10]LCD display: Pressure digital(accuracy:0.001 Mpa), Accumulating timing(range:0-99999 Hours), Present timing(accuracy:1 Minute), Presetting timing(accuracy: 1 Minute), Optional : Temperature digital(accuracy:0.1);
11]Alarm: Low purity
12]Standard Accessories: Oxygen Concentrator, Oxygen tube, Humidifier bottle, Primary air suction filtering sponge, Secondary air suction filters, General Machine ManualWarranty Period:
The product from date of sale, under normal use and storage conditions, the warranty period of this product is 24 months or 5000 Hrs (whichever is the earliest).
The following circumstances shall not be covered within the free warranty of the company:
1]Damage due to improper operation of users or use under abnormal conditions;
2]Damage deformation of machine (including components and parts) due to collision or falling;
3]Damage due to dis-assembly, repair or modification by users;
4]Damage caused by natural disasters;
5]Wearing parts and consumables: atomizer, filtering foam, filtering cotton etc.